1. Hambly K, Sibley R, Bs MOM. Level of agreement between a novel smartphone application and a long arm goniometer for the assessment of maximum active knee flexion by an inexperienced tester. :15.
2. Castle H, Kozak K, Sidhu A, Khan RJK, Haebich S, Bowden V, et al. Smartphone technology: a reliable and valid measure of knee movement in knee replacement. Int J Rehabil Res. 2018 Jun;41(2):152–8.
3. Alawna MA, Unver BH, Yuksel EO. The Reliability of a Smartphone Goniometer Application Compared With a Traditional Goniometer for Measuring Ankle Joint Range of Motion. J Am Podiatr Med Assoc. 2019 Jan 1;109(1):22–9.
4. Shamsi M, Mirzaei M, Khabiri SS. Universal goniometer and electro-goniometer intra-examiner reliability in measuring the knee range of motion during active knee extension test in patients with chronic low back pain with short hamstring muscle. BMC Sports Sci Med Rehabil. 2019 Dec;11(1):4.
5. Ishii K, Oka H, Honda Y, Oguro D, Konno Y, Kumeta K, et al. Accuracy and reliability of a smartphone application for measuring the knee joint angle. J Phys Ther Sci. 2021;33(5):417–22.
6. Derhon V, Santos RA, Brandalize M, Brandalize D, Rossi LP. Intra- and Inter-Examiner Reliability in Angular Measurements of the Knee with a Smartphone Application. Hum Mov [Internet]. 2017 Jan 1 [cited 2021 Jun 5];18(2). Available from: https://www.termedia.pl/Intra-and-inter-examiner-reliability-in-angular-measurements-of-the-knee-with-a-smartphone-application,129,31947,0,1.html
7. Jones A, Sealey R, Crowe M, Gordon S. Concurrent validity and reliability of the Simple Goniometer iPhone app compared with the Universal Goniometer. Physiother Theory Pract. 2014 Oct;30(7):512–6.
8. Johnson LB, Sumner S, Duong T, Yan P, Bajcsy R, Abresch RT, et al. Validity and reliability of smartphone magnetometer-based goniometer evaluation of shoulder abduction – A pilot study. Man Ther. 2015 Dec;20(6):777–82.
9. Rodríguez-Sanz J, Carrasco-Uribarren A, Cabanillas-Barea S, Hidalgo-García C, Fanlo-Mazas P, Lucha-López MO, et al. Validity and reliability of two Smartphone applications to measure the lower and upper cervical spine range of motion in subjects with chronic cervical pain. J Back Musculoskelet Rehabil. 2019 Jul 23;32(4):619–27.
10. Dos Santos RA, Derhon V, Brandalize M, Brandalize D, Rossi LP. Evaluation of knee range of motion: Correlation between measurements using a universal goniometer and a smartphone goniometric application. J Bodyw Mov Ther. 2017 Jul;21(3):699–703.
11. Werner BC, Holzgrefe RE, Griffin JW, Lyons ML, Cosgrove CT, Hart JM, et al. Validation of an innovative method of shoulder range-of-motion measurement using a smartphone clinometer application. J Shoulder Elbow Surg. 2014 Nov;23(11):e275–82.
12. Milanese S, Gordon S, Buettner P, Flavell C, Ruston S, Coe D, et al. Reliability and concurrent validity of knee angle measurement: Smartphone app versus universal goniometer used by experienced and novice clinicians. Man Ther. 2014 Dec;19(6):569–74.
13. Brosseau L, Balmer S, Tousignant M, O’Sullivan JP, Goudreault C, Goudreault M, et al. Intra- and intertester reliability and criterion validity of the parallelogram and universal goniometers for measuring maximum active knee flexion and extension of patients with knee restrictions. Arch Phys Med Rehabil. 2001 Mar;82(3):396–402.
14. Cooper G, Sheret I, McMillian L, Siliverdis K, Sha N, Hodgins D, et al. Inertial sensor-based knee flexion/extension angle estimation. J Biomech. 2009 Dec;42(16):2678–85.
15. Ockendon M, Gilbert R. Validation of a Novel Smartphone Accelerometer-Based Knee Goniometer. J Knee Surg. 2012 May 3;25(04):341–6.